Ultrasonic Flowmeter - Probe Installation
Ultrasonic flowmeter must be installed in the installation position, so that it can better prevent the installation of ultrasonic flowmeter problems, then the installation location should avoid what?
1 , to avoid the installation of the machine in the pump, high-power radio, frequency conversion, that is, a strong magnetic field and vibration interference ;
2. Select pipe sections that should be uniformly dense and easy to transmit ultrasonic waves ;
3 , must have a long enough straight section, the upstream straight section of the installation point must be greater than 10D ( Note: D = diameter ) , the downstream should be greater than 5D;
4 , the installation point upstream from the pump should have 30D distance ;
5 , the fluid should be filled with pipes ;
Ultrasonic flow meter installation requirements are simple and convenient in the installation of all flow meters. Simply select a suitable measurement point, input the piping parameters at the measurement point into the flow meter, and then fix the probe on the pipe.
Perhaps you are interested in the company's orifice flowmeter , electromagnetic flowmeter , pressure transmitter , paperless recorder , welcome new and old customers to consult.
Antimony Concentrate,Gold Concentrate,Ore Of Antimony,Antimony And Gold Raw Materials
HUNAN ZHONGNAN ANTIMONY&TUNGSTEN TRADING CO.,LTD , https://www.znat.com.cn